X2SR01 Astra XT SR MRI SureScan
Implantable Pulse Generator (IPG)
Device Survival Probability
- Including Normal Battery Depletion – This curve includes devices that have reached at least 80% of expected longevity. This curve is most representative of clinical performance and how long the device will last.
- Excluding Normal Battery Depletion – This is the malfunction free survival curve.
Customer Communications For This Model
|Max. Delivered Energy
Hide this content
|Serial Number Prefix
||500 Lead Ω
||1000 Lead Ω
|Low 2.5 V (A, RV)
|Nominal 3.5 V (A, RV)
|High 5.0 V (A, RV)
From the point that the RRT is set, the pacemaker will operate for approximately six months for typical pacemaker configurations during the normal operating life.
Recommended Replacement (RRT)
Battery voltage less than or equal to 2.63V on 3 consecutive daily automatic measurements.
Data as of April 10, 2020