A5DR01 Advisa DR
Implantable Pulse Generator (IPG)
Advisories For This Model
- There are no advisories for this model.
Performance Notes For This Model
- There are no performance notes for this model.
|Max. Delivered Energy
Hide this content
|Serial Number Prefix
||500 Lead Ω
||1000 Lead Ω
|Low 2.5 V (A, RV)
|Nominal 3.5 V (A, RV)
|High 5.0 V (A, RV)
The Prolonged Service Period (PSP) is the time between the RRT and EOS. The PSP is defined as 6 months assuming the following conditions: 100% VVI pacing at 60 min–1, 2.5 V RV pacing amplitude; 0.4 ms pulse width; and 600 Ω pacing load. The EOS may be indicated before the end of 6 months if the device exceeds these conditions.
Recommended Replacement (RRT)
Battery voltage less than or equal to 2.83V on 3 consecutive daily automatic measurments.
Data as of October 12, 2018